The KODAK Intraoral X-ray system is equipped with a thermal safety system that prevents the generator from overheating in case of intensive use. Product, Model Kodak Intraoral X-Ray System, Catalog Numbers , , , XR — Carestream Health, Inc. Made in France Kodak Intraoral X-Ray System Performance. Operational ease. Safety. The Kodak intraoral x-ray system is designed for performance and safety.
|Published (Last):||18 April 2011|
|PDF File Size:||3.73 Mb|
|ePub File Size:||15.93 Mb|
|Price:||Free* [*Free Regsitration Required]|
KODAK intraoral X-ray system is unveiled :Dental Tribune USA
Chad Gehani, an endodontist, as its new leader. Complete control The Kodak system is equipped with adjustable tension 60 or 70 kV to accentuate contrast and levels of gray shading according to your diagnostic needs. The resistor is very fragile!
Please Mr Faustino can you send me the service manual and im interessed to inyour spare parts can you mail me suissedental gmail. December 13, New law seeks to improve access to oral health care intrxoral underserved patients.
Kodak 2200 Kv_error
Open the catalog to page 2. Related Searches Flat panel detector Dental imaging software Medical radiology software Cephalometric analysis software Dental software Simulation software Analysis software. It shows no numbers just KV Error.
This advanced control helps you obtain optimized images that are just the way you want them—giving you total control over your settings. Avoid Coding Errors 12 Feb Intraorral excessive force is applied, then the ceramic resistor will break in pieces! Equipped with a focal spot of 0. And, after each exposure, the dose level is displayed for simple dose monitoring.
KODAK 2200 Intraoral X-Ray System
The voltage devider must be immersed in oil completely, no air bubbles present in oil! And with multiple installation configurations available, you can easily adapt the unit to match the needs of your practice. Optimized for both film and digital radiography usage, the system includes a redesigned electronic timer that makes the setting of exposure parameters failsafe.
I drill a 1 mm hole near anode’s contact in the plastic shell of the tube to be sure that all air bubbles are eliminated. Registration Forgot your password?
Thanks, I will forward your email for those who need help on tube repairs and software. Call Email DOTmedinfo oxinst. I have the same problem Can anybody send me the service manual for please my email address: Can be located anywhere from close to the tube to outside of the operating room Timer can also be mounted on the wall-framework. You need the K control panel to service the unit.
Kodak Intraoral X-Ray System – IMAGELEVEL – PDF Catalogs | Technical Documentation
My panel is without rotating timer knob. I knew can’t find it anywhere. I opened the tube collect all that oil.
My presentations Profile Feedback Log out. The soldering on the resistor gets loose and this is the reason why so many of you get error I would recommend you not dismantle the ceramic resistor.
For next procedure for self-calibration: Multiple configurations to fit any practice Compatible with both film and digital imaging applications, the Kodak system works the way you do.
Open the catalog to page 1. koadk
Advertise Increase Your Brand Awareness. With the new system timer, dental professionals can quickly and easily select the most appropriate exposure settings based on the teeth to be examined, the patient size and receptor used.
Hi, KV error on a means either a power board failure or a tubehead failure which is the most frequent problem. Just check if it’s ok 13 Kohm and MOhn and solder unplagged wire to it. Please, I need the service manual. The best regsrds, Alex.