High speed – up to 15,rpm/21, x g * Whisper quiet brushless induction drive * Precise, microprocessor control * Automatic g-force conversion program. Hermle ZM-2 Microcentrifuge from Labnet International, Inc. A high performance microlitre centrifuge, offering a fullrange of features:digital display of time, rpm and RCFtwo acceleration and deceleration.
|Published (Last):||15 November 2016|
|PDF File Size:||20.25 Mb|
|ePub File Size:||15.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
There is a distinction between two different kinds of errors. Enter text from picture: Keyboard – Starting The Centrifuge – “quick”-key 4. Don’t show me this message again.
Hermle z m-2 Manuals
Preselection Of Operating Time 4. Page 14 – Keyboard – Starting the centrifuge – “qu Don’t have an account? Page 20 Error no.: Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the snap Before operation, secure the rotor lid to the rotor by pressing the heermle Description – The rotor is not installed securely on the motor shaft.
General Information The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations.
To open the centrifuge lid. The temperature rise depends on the rotor swing-out or angle rotor ,bucket type, ambient temperature, running time and the speed of the rotor. Width 28 cm Depth 37 cm Height Service And Inspection Of Centrifuge 7. Operation Operation Operation Operation Operation 4. Figure 8 If the green “fast” control lamp is lit, the unit accelerates and decelerates fast.
Power failure while rotor is in motion Action: The Hermle Z M-2 is neither explosion proof nor inert gas shielded and should, therefore, never be operated in explosive-hazardous locations. The digital indications on the display will light up when the power is on.
Comments to this Manuals Your Name. Open herm,e reclose the lid, restart the centrifuge Error no.: The green control lamps 3 show which mode is activated. Commercially available tubes or bottles made of glass or plastic. her,le
Page 15 – Temperture Page 16 – Service and inspection of centrifuge Page 17 – Cleaning of centrifuge after breakage of Turn the speed knob 1see figure 4 till the required rotor code appears on the d i s p l a y.
To print hernle manual completely, please, download it. Page of 20 Go. Got it, continue to print. On the digital display “time min-sec” the letters “ro” rotor appears. By pressing ‘print’ button you will print only current page. Never stay in the safety zone of 30 cm around the centrifuge or deposit dangerous materials inside this zone during centrifugation. Use only original spare parts!
Centrifuges Z 233 M-2 -Hermle- Germany
hermlee Overcurrent or undervoltage due to power supply fluctuations Action If this is not done, they could scratch the protective coating of the rotor. The rotor must always be balanced in this fashion prior to operation.
Table of Contents Add to my manuals Add. Page 19 Error no.: Should this happen, the centrifuge, rotor, buckets, adapters and the rotor chamber must be thoroughly cleaned and all broken particles removed immediately. With the centrifuge lid open you can preset the operating time with the knob 3. Set Or Change Rotor-code -Load the rotor symmetrically by inserting tubes into spaces opposite from each other.
During a run and with the centrifuge lid closed you have to hermlee the herml key 1hold it and change operating time by turning the knob 3.
If the green “slow”control lamp is lit, the unit accelerates and decelerates very softly below 2 rpm.